From: Frida on 9 May 2006 05:40
I am working with the HP 34970A instrument togehter with labview. I have a VI "HP34970A Conf Trigger.vi" that comes with the toolkit I can download from your site. But the description/help for the VI is a bit poor. It says I can use this VI to control the scan interval for example. But I can seem to get it work. This Vi has 9 parameter inputs and I have tried some differnet combinations (the most resonable, I thought), but none of them is right (apperently). I would like to be able to control the scan interval from my main VI through a shared variable.
Does anyone have any idea how to configure this VI to make the scan controlling work??I'll be glad for any help!
From: Jimmie A. on 10 May 2006 06:10
Dennis Knutson gave you some good tips and suggestions!
Down below you will find some additional information that might be helpful:
A tutorial that can be good to read
Using the Agilent 34970A with LabVIEW
<a href="http://zone.ni.com/devzone/conceptd.nsf/webmain/166200887455792786256fe3006901ae" target="_blank"> http://zone.ni.com/devzone/conceptd.nsf/webmain/166200887455792786256fe3006901ae</a>
An example that can be helpful
Agilent Data Acquisition system 34970A
<a href="http://sine.ni.com/apps/we/niepd_web_display.display_epd4?p_guid=E67ED8E9BA553F06E0340003BA230ECF&p_node=DZ52282&p_source=External" target="_blank"> http://sine.ni.com/apps/we/niepd_web_display.display_epd4?p_guid=E67ED8E9BA553F06E0340003BA230ECF&p_node=DZ52282&p_source=External</a>